For research use only. Not for use in diagnostic procedures
1) If flow stops and the temperature remains on, then buffer can crystallizes in probe tip causing them to clog.
2) If finger tight union is over tightened, it can crush the capillary. Broken silica particles can travel to tip of capillary and clog electrode.