For research use only. Not for use in diagnostic procedures.
If the positive and negative reference table fields are empty after opening Analyst® software, the reference table can be restored by following these steps:
- Back up the API Instrument folder
- Open the backup of the API Instrument\Preferences folder
- Copy the TuneData.tun file over to the current API instrument folder/Preferences folder
The calibration table should be restored.