For research use only. Not for use in diagnostic procedures.
Analyst® software activates and launches acquisition methods successfully with MS only hardware profiles. When a Shimadzu LC is added to the hardware profile, the software crashes.
If the Shimadzu LC system has a rack controller, it must be checked on the autosampler portion of the hardware profile. Use the following steps to add the rack controller:
- Within the Hardware Profile select Shimadzu LC Controller and then select Setup Device.
- In the Shimadzu Configuration screen, select Configure.
- In the Shimadzu HPLC System Configuration screen (see image below), select Configure next to Autosampler.
- In the Autosampler Configuration screen (see image below), select the check box next to the Rack Changer Installed option.