ProteinPilot™ software uses the incorrect mass spectrometer name in its group file output file. When a TripleTOF® 5600 system was selected, the group file contains the name QSTAR® Elite instead for system information. This causes the wrong results to be output; if the incorrect instrument name is used, then the incorrect tolerance settings are applied.
An error results when selecting the ACQ-BSM tab when creating an LC method using the Acquity Method Editor accessed through Analyst® software. It can be resolved by selecting any other tab in the method editor screen.
Upgrading from Analyst® 1.6.3 to Analyst 1.7 software in a Windows 10 environment will cause an installation error, and Analyst 1.7 software will not launch. Analyst 1.7 with Hotfix 1 must be installed alone, and previous versions of Analyst software cannot also be installed concurrently.
When installing Analyst® 1.7 software with Hotfix 1 and the patch for Analyst 1.7 for Windows 10 Update 1709, the following error message was received, "You must have Analyst 1.7 software installed to install Analyst 1.7 Patch for Windows 10 Update 1709."
If the computer clock updates for daylight savings time while MPX™ software is submitting a batch, data corruption and batch stoppage can occur. The clock must be set up so that automatic updates to the time do not occur.