For research use only. Not for use in diagnostic procedures.
If an Exion LC 2.0 system contains a DAD or multiwavelength detector, then after the LC system and mass spectrometer go to Standby state, the Analyst® software status icon for the ExionLC 2.0 system turns yellow, but the LC system state is shown as Wait. This occurs because the detector lamps are turned off when the system is in Standby state. This is a status display issue and does not affect system operation.